Check us out at http://www.tutorvista.com//videos A Wheatstone bridge is a measuring instrument invented by Samuel Hunter Christie in 1833 and improved and popularized by Sir Charles Wheatstone in 1843 . It is used to measure an unknown electrical resistance by balancing two legs of a bridge circuit, one leg of which includes the unknown component. Its operation is similar to the original potentiometer. Rx is the unknown resistance to be measured; R1, R2 and R3 are resistors of known resistance and the resistance of R2 is adjustable. If the ratio of the two resistances in the known leg (R2 / R1) is equal to the ratio of the two in the unknown leg (Rx / R3), then the voltage between the two midpoints (B and D) will be zero and no current will flow through the galvanometer Vg. R2 is varied until this condition is reached. The direction of the current indicates whether R2 is too high or too low. Detecting zero current can be done to extremely high accuracy. Therefore, if R1, R2 and R3 are known to high precision, then Rx can be measured to high precision. Very small changes in Rx disrupt the balance and are readily detected. At the point of balance, the ratio of R2 / R1 = Rx / R3 Therefore, Alternatively, if R1, R2, and R3 are known, but R2 is not adjustable, the voltage difference across or current flow through the meter can be used to calculate the value of Rx, using Kirchhoff's circuit laws (also known as Kirchhoff's rules). This setup is frequently used in strain gauge and resistance thermometer measurements, as it is usually faster to read a voltage level off a meter than to adjust a resistance to zero the voltage.
Questions about Wheatstone Bridge
Want more info about Wheatstone Bridge?
Get free advice from education experts and Noodle community members.